REDLAB 333 – Laboratorio FIDAMC
DSC TA Q2000
Differential scanning calorimetry (DSC) is a thermal analysis technique which measures the temperature and heat flow associated with transitions in materials as a function of temperature and time. Determination of the glass transition temperature, cure degree, heat capacity and study of the cure reaction of thermoset matrix composite materials.
- Temperature range: -90 – 550 ºC
- Heating/cooling rate: máx 100 ºC/min
- Precision: +/-0.01 ºC
- Calorimetric precision: +/-0.05 %
Laboratory de FIDAMC
DMA TA Q800
Dynamic mechanical analysis (DMA) is a thermoanalytical technique used for studying the viscoelastic nature of polymers. Determination of the glass transition temperature and viscoelastic properties of polymers and polymer composites.
- Temperature range: -145 – 600 ºC
- Heating rate: 0.1 – 20 ºC/min
- Modulus: range 1 kPa – 3 TPa, precision +/-1 %
Laboratory of FIDAMC
Universal Testing machine
MTS Landmark Sistema de Ensayo Servohidraulico mod. 370.1
Universal mechanical testing machine (static and dynamic) calibrated to tensile and compression. Performing a variety of static tests (Compression, IPSS, Tensile, core tests …) and dynamic tests (fatigue).
The load capacity of the machine is 5 and 100 kN, the working temperature range is -120 ° C to 315 ° C since a thermal chamber can be coupled by having two types of jaws.
Mechanical laboratory of Fidamc (Tecnogetafe).
Cutting machine Brillant 285
Automatic cutting machine with refrigerated disc. Cutting and preparation of panels to obtain samples.
The maximum dimensions of panels: 500×700 mm. The diamond disc for cutting can be up to 600mm in diameter.
Specimen preparation laboratory of Fidamc (Tecnogetafe).
Manual Ultrasound flaw detector: OLYMPUS EPOCH 650
Ultrasound interrogator for NDT analysis to ensure composite structural integrity.
- Pulse-echo analysis
- 1-10 MHz bandwith
- 1/4″ probe diameter
- PRF up to 2kHz
FIDAMC workshop (Tecnogetafe).